DIN standard for the determination of the tip radius and the probing force of stylus instruments developed
In microsystems metallic components are increasingly being replaced by those from low-cost polymers. For the thickness measurement of polymers, there is now the DIN standard 32567 available, which describes both, optical and tactile surface measuring methods for the precise measurement of the thickness of polymer layers.
The standard describes methods by which both, the contact force and the tip radius of stylus instruments can be measured, a basic condition for non-destructive precision tactile profile measurements.
From motion sensor to smartphones - many everyday products increasingly contain parts which are made of polymer materials. The desired operation of these components depends not only on the dimensions, often also on the mechanical properties of these materials.
The dimensions can be measured optically or tactilely. However, in the thickness measurement of transparent materials with optical measuring methods, but also in the stylus measurement of coatings on hard substrates systematic deviations of the measured thickness are observed. In tactile methods, the main influencing factors are the probing force and the tip radius.
With viscous materials whose mechanical properties are time-dependent, also different scanning speeds affect the level of systematic deviations. PTB has therefore, in cooperation with other European national metrology institutes, developed a method for correcting these systematic errors, which has been standardized in DIN 32567. In the standard, the main influencing factors are shown for tactile and optical measurements and methods for the estimation, correction and reduction of systematic errors are described.
Contact at PTB
Dr. Uwe Brand, Working Group 5.11 Hardness and Tactile Probing Methods, Telefon: 0531-592 5111, E-Mail: email@example.com
• Brand, U.; Beckert, E.; Beutler, A.; Dai, G.; Stelzer, C.; Hertwig, A.; Klapetek, P.; Koglin, J.; Thelen, R. and Tutsch, R.: Comparison of optical and tactile layer thickness measurements of polymers and metals on silicon or SiO2. Meas. Sci. Technol. 22 (2011) 094021 (14pp)
• DIN 32567 Fertigungsmittel für Mikrosysteme — Ermittlung von Materialeinflüssen auf die Messunsicherheit in der optischen und taktilen dimensionellen Messtechnik. Teile 1 – 5
• Li, Z., Brand, U. und Ahbe, T.: Step height measurement of microscale thermoplastic polymer specimens using contact stylus profilometry. Prec. Eng. 45, 110–117 (2016)
Imke Frischmuth | Physikalisch-Technische Bundesanstalt (PTB)
Researchers demonstrate existence of new form of electronic matter
15.03.2018 | University of Illinois at Urbana-Champaign
Boron can form a purely honeycomb, graphene-like 2-D structure
15.03.2018 | Science China Press
Animal photoreceptors capture light with photopigments. Researchers from the University of Göttingen have now discovered that these photopigments fulfill an...
On 15 March, the AWI research aeroplane Polar 5 will depart for Greenland. Concentrating on the furthest northeast region of the island, an international team...
The world’s second-largest ice shelf was the destination for a Polarstern expedition that ended in Punta Arenas, Chile on 14th March 2018. Oceanographers from...
At the 2018 ILA Berlin Air Show from April 25–29, the Fraunhofer Institute for Laser Technology ILT is showcasing extreme high-speed Laser Material Deposition (EHLA): A video documents how for metal components that are highly loaded, EHLA has already proved itself as an alternative to hard chrome plating, which is now allowed only under special conditions.
When the EU restricted the use of hexavalent chromium compounds to special applications requiring authorization, the move prompted a rethink in the surface...
At the ILA Berlin, hall 4, booth 202, Fraunhofer FHR will present two radar sensors for navigation support of drones. The sensors are valuable components in the implementation of autonomous flying drones: they function as obstacle detectors to prevent collisions. Radar sensors also operate reliably in restricted visibility, e.g. in foggy or dusty conditions. Due to their ability to measure distances with high precision, the radar sensors can also be used as altimeters when other sources of information such as barometers or GPS are not available or cannot operate optimally.
Drones play an increasingly important role in the area of logistics and services. Well-known logistic companies place great hope in these compact, aerial...
16.03.2018 | Event News
13.03.2018 | Event News
08.03.2018 | Event News
16.03.2018 | Earth Sciences
16.03.2018 | Physics and Astronomy
16.03.2018 | Life Sciences