Portable radiation detectors generally perform well enough to meet new consensus standards but provide inaccurate readings for certain types of radiation, according to recent tests by the National Institute of Standards and Technology (NIST).
The results, reported in the May issue of the journal Health Physics,* are based on NIST tests of 31 commercial detectors, including hand-held survey meters; electronic personal alarming detectors (similar to pagers); and radionuclide identifiers (specialized devices that can identify specific radioactive materials). A number of federal, state and local agencies are using such instruments as part of homeland security-related efforts to detect and identify radioactive materials.
Researchers compared the devices exposure rate readings to NIST measurements for different energy and intensity levels produced by NISTs calibrated gamma ray and X-ray beam lines. The responses of the majority of the detectors agreed with NIST-measured values, within acceptable uncertainties, for tests with gamma rays. This performance meets requirements established by new American National Standards Institute (ANSI) standards, adopted by the Department of Homeland Security (DHS) in 2004. However, there was a large discrepancy between most detectors readings and the NIST values for the lowest-energy X-rays. For instance, readings by 14 detectors were roughly 40 to 100 percent below the NIST value. The deviations were much larger than those stated in manufacturers specifications.
Laura Ost | EurekAlert!
Smallest transistor worldwide switches current with a single atom in solid electrolyte
17.08.2018 | Karlsruher Institut für Technologie (KIT)
Protecting the power grid: Advanced plasma switch for more efficient transmission
17.08.2018 | DOE/Princeton Plasma Physics Laboratory
New design tool automatically creates nanostructure 3D-print templates for user-given colors
Scientists present work at prestigious SIGGRAPH conference
Most of the objects we see are colored by pigments, but using pigments has disadvantages: such colors can fade, industrial pigments are often toxic, and...
Scientists at the University of California, Los Angeles present new research on a curious cosmic phenomenon known as "whistlers" -- very low frequency packets...
Scientists develop first tool to use machine learning methods to compute flow around interactively designable 3D objects. Tool will be presented at this year’s prestigious SIGGRAPH conference.
When engineers or designers want to test the aerodynamic properties of the newly designed shape of a car, airplane, or other object, they would normally model...
Researchers from TU Graz and their industry partners have unveiled a world first: the prototype of a robot-controlled, high-speed combined charging system (CCS) for electric vehicles that enables series charging of cars in various parking positions.
Global demand for electric vehicles is forecast to rise sharply: by 2025, the number of new vehicle registrations is expected to reach 25 million per year....
Proteins must be folded correctly to fulfill their molecular functions in cells. Molecular assistants called chaperones help proteins exploit their inbuilt folding potential and reach the correct three-dimensional structure. Researchers at the Max Planck Institute of Biochemistry (MPIB) have demonstrated that actin, the most abundant protein in higher developed cells, does not have the inbuilt potential to fold and instead requires special assistance to fold into its active state. The chaperone TRiC uses a previously undescribed mechanism to perform actin folding. The study was recently published in the journal Cell.
Actin is the most abundant protein in highly developed cells and has diverse functions in processes like cell stabilization, cell division and muscle...
17.08.2018 | Event News
08.08.2018 | Event News
27.07.2018 | Event News
17.08.2018 | Physics and Astronomy
17.08.2018 | Information Technology
17.08.2018 | Life Sciences