The dosemeters used to measure beta radiation have been subjected to an annual quality control by the PTB for six years now. Since then, all dosimeter models used in Germany have passed these controls.
In Germany, for persons who are occupationally exposed to radiation, monitoring with personal dose equivalent meters is legally prescribed. Besides the whole-body dosimeters which are to be worn on the trunk, also passive extremity dosimeters are used which are often worn around the fingers. Extremity dosimeters are used for beta radiation measurements, especially in medicine. All dosimeters of this type are released and assessed by dosimetry services. PTB is thereby responsible for quality control. Quality control takes place in two stages:
In the first stage, the designs are submitted to a thorough check with regard to their properties before the dosimeter is released for use. In the case of beta dosimeters, this control is carried out according to a directive.
In the second stage, the routine operation is checked yearly at the dosimetry services via so-called "periodic comparison measurements". The dosimeters are thereby exposed by PTB to a well known dose - which we we will call HPTB in the following. Then, the dosimetry services analyse the dosimeters and determine their measured value, HDST, of course without knowing the value of HPTB. As a measure of quality, the quotient HDST/HPTB is used which is 1.0 for an ideal dosimeter. The figure shows the values of these quotients in relation with the dose for all comparison measurements which have been carried out with beta radiation up to now. The lines plotted indicate the admissible limits for this quotient. Due to the requirements of radiation protection, these limits narrow towards high doses.
The condition for passing the yearly comparison measurements is that 90% of all quotients for each design of a service lie within these limits. If this requirement is not met by one dosimeter design for the comparison measurement, a repetition is carried out for this dosimeter. This has never been the case up to now. In summary, one can say that this type of two-stage quality control is particularly efficient.Contact:
Erika Schow | alfa
Explosion on Jupiter-sized star 10 times more powerful than ever seen on our sun
18.04.2019 | University of Warwick
In vivo super-resolution photoacoustic computed tomography by localization of single dyed droplets
18.04.2019 | Changchun Institute of Optics, Fine Mechanics and Physics, Chinese Academy of Sciences
A stellar flare 10 times more powerful than anything seen on our sun has burst from an ultracool star almost the same size as Jupiter
A localization phenomenon boosts the accuracy of solving quantum many-body problems with quantum computers which are otherwise challenging for conventional computers. This brings such digital quantum simulation within reach on quantum devices available today.
Quantum computers promise to solve certain computational problems exponentially faster than any classical machine. “A particularly promising application is the...
The technology could revolutionize how information travels through data centers and artificial intelligence networks
Engineers at the University of California, Berkeley have built a new photonic switch that can control the direction of light passing through optical fibers...
Physicists observe how electron-hole pairs drift apart at ultrafast speed, but still remain strongly bound.
Modern electronics relies on ultrafast charge motion on ever shorter length scales. Physicists from Regensburg and Gothenburg have now succeeded in resolving a...
Engineers create novel optical devices, including a moth eye-inspired omnidirectional microwave antenna
A team of engineers at Tufts University has developed a series of 3D printed metamaterials with unique microwave or optical properties that go beyond what is...
17.04.2019 | Event News
15.04.2019 | Event News
09.04.2019 | Event News
18.04.2019 | Life Sciences
18.04.2019 | Physics and Astronomy
18.04.2019 | Life Sciences