Wearable cardioverter defibrillators are used by people who may be at higher risk for sudden cardiac arrest, including those with weakened heart function, awaiting cardiac transplant or with a condition that prevents or delays them from receiving an implanted defibrillator.
The device monitors heart rhythm, emits alarms if a serious arrhythmia occurs, delivers an electric shock to the heart if needed and alerts bystanders to help if the heart's electrical activity has stopped.
About 5,000 patients are using wearable defibrillators at any one time, usually for about 60 days, said Vincent N. Mosesso Jr., M.D., professor of emergency medicine at the University of Pittsburgh School of Medicine and principal investigator of the study.
"In these patients, the wearable defibrillator is a non-invasive 'insurance policy' against sudden arrest during their vulnerable period," he said.
Researchers gathered heart rhythm records and calls about shocks from a registry of 14,475 patients with wearable defibrillators listed from 2007 through 2009. Of those, 185 (about 1 percent) received an appropriate shock and 91.6 percent survived one or more episodes of ventricular fibrillation or ventricular tachycardia, the most common abnormal rhythms during cardiac arrest.
Wearable defibrillators delivered 223 inappropriate shocks to 213 people who weren't experiencing ventricular fibrillation or ventricular tachycardia. However, no one died as a result. Researchers attribute the inappropriate shocks to signal noise, rapid non-VT rhythms, and rhythm misinterpretation.
Only about 7 percent of people in the United States who have sudden cardiac arrest outside the hospital survive to hospital discharge, and only about 21 percent who have them in the hospital survive to discharge.
"This study confirms the effectiveness of very early defibrillation as therapy for sudden cardiac arrest in high-risk patients when delivered by a wearable defibrillator," Mosesso said. "These defibrillators provide patients the critical advantage of not having to wait for a bystander or emergency responder to recognize the cardiac arrest and use an automated external defibrillator or manual defibrillator — both of which can lead to delays in treatment and markedly worse survival rates."
Co-authors are: Jie Li, M.S.; Douglas Landsittel, Ph.D. and Leonard I. Ganz, M.D. Author disclosures are on the abstract.
ZOLL LifeCor of Pittsburgh, which makes the wearable defibrillator used in the study, funded the research.
Statements and conclusions of study authors that are presented at American Heart Association scientific meetings are solely those of the study authors and do not necessarily reflect association policy or position. The association makes no representation or warranty as to their accuracy or reliability. The association receives funding primarily from individuals; foundations and corporations (including pharmaceutical, device manufacturers and other companies) also make donations and fund specific association programs and events. The association has strict policies to prevent these relationships from influencing the science content. Revenues from pharmaceutical and device corporations are available at www.heart.org/corporatefunding.
AHA News Media Office | EurekAlert!
Can radar replace stethoscopes?
14.08.2018 | Friedrich-Alexander-Universität Erlangen-Nürnberg
Novel PET imaging method could track and guide therapy for type 1 diabetes
03.08.2018 | Society of Nuclear Medicine and Molecular Imaging
There are currently great hopes for solid-state batteries. They contain no liquid parts that could leak or catch fire. For this reason, they do not require cooling and are considered to be much safer, more reliable, and longer lasting than traditional lithium-ion batteries. Jülich scientists have now introduced a new concept that allows currents up to ten times greater during charging and discharging than previously described in the literature. The improvement was achieved by a “clever” choice of materials with a focus on consistently good compatibility. All components were made from phosphate compounds, which are well matched both chemically and mechanically.
The low current is considered one of the biggest hurdles in the development of solid-state batteries. It is the reason why the batteries take a relatively long...
New design tool automatically creates nanostructure 3D-print templates for user-given colors
Scientists present work at prestigious SIGGRAPH conference
Most of the objects we see are colored by pigments, but using pigments has disadvantages: such colors can fade, industrial pigments are often toxic, and...
Scientists at the University of California, Los Angeles present new research on a curious cosmic phenomenon known as "whistlers" -- very low frequency packets...
Scientists develop first tool to use machine learning methods to compute flow around interactively designable 3D objects. Tool will be presented at this year’s prestigious SIGGRAPH conference.
When engineers or designers want to test the aerodynamic properties of the newly designed shape of a car, airplane, or other object, they would normally model...
Researchers from TU Graz and their industry partners have unveiled a world first: the prototype of a robot-controlled, high-speed combined charging system (CCS) for electric vehicles that enables series charging of cars in various parking positions.
Global demand for electric vehicles is forecast to rise sharply: by 2025, the number of new vehicle registrations is expected to reach 25 million per year....
17.08.2018 | Event News
08.08.2018 | Event News
27.07.2018 | Event News
20.08.2018 | Information Technology
20.08.2018 | Life Sciences
20.08.2018 | Information Technology