Since it works with samples close to a thousand times smaller than comparable commercial instruments, the new technique should be an important addition to the growing arsenal of measurement tools for nanotechnology, according to the NIST team.
As the objects of scientific research have gotten smaller and smaller—as in nanotechnology and gene therapy—the people who worry about how to measure these things have been applying considerable ingenuity to develop comparable instrumentation.* This new NIST technique is a riff on thermogravimetric analysis (TGA), an imposing name for a fairly straightforward concept. A sample of material is heated, very slowly and carefully, and changes in its mass are measured as the temperature increases. The technique measures the reaction energy needed to decompose, oxidize, dehydrate, or otherwise chemically change the sample with heat.
TGA can be used, for example, to characterize complex biofuel mixtures because the various components vaporize at different temperatures. The purity of an organic sample can be tested by the shape of a TGA plot because, again, different components will break down or vaporize at different temperatures. Conventional TGA, however, requires samples of several milligrams or more of material, which makes it hard to measure very small, laboratory-scale powder samples—such as nanoparticles—or very small surface chemistry features such as thin films.
What's needed is an extremely sensitive "microbalance" to measure the minute changes in mass. The NIST group found one in the quartz crystal microbalance, essentially a small piezoelectric disk of quartz sandwiched between two electrodes. An alternating current across the electrodes causes the crystal to vibrate at a stable and precise ultrasonic frequency—the same principle as a quartz crystal watch. Added mass (a microsample) lowers the resonant frequency, which climbs back up as the microsample is heated and breaks down.
In a new paper.** the NIST materials science group demonstrates that their microbalance TGA produces essentially the same results as a conventional TGA instrument, but with samples about a thousand times smaller. They can detect not only the characteristic curves for carbon black, aluminum oxide and a sample organic fluid, but also the more complex curves of mixtures.
"We started this work because we wanted to analyze the purity of small carbon nanotube samples," explains analytical chemist Elisabeth Mansfield. More recently, she says, they've applied the technique to measuring the organic surface coatings biologists put on gold nanoparticles to modify them for particular applications. "Measuring how much material coats the particles surface is very hard to do right now," she says, "It will be a really unique application for this technique."
The prototype apparatus requires that the frequency measurements be made in a separate step from the heating. Currently, the team is at work integrating the microbalance disks with a heating element to enable the process to be simultaneous.
* See, for example, "Micro Rheometer is Latest Lab On a Chip Device" in NIST Tech Beat, Aug. 31, 2010, www.nist.gov/public_affairs/tech-beat/tb20100831.cfm#rheometer.
** E. Mansfield, A. Kar, T.P. Quinn and S.A. Hooker. Quartz crystal microbalances for microscale thermogravimetric analysis. Anal. Chem. Article ASAP, published online Nov. 16, 2010. DOI: 10.1021/ac102030z
Michael Baum | EurekAlert!
Nanobot pumps destroy nerve agents
21.08.2018 | American Chemical Society
How do muscles know what time it is?
21.08.2018 | Helmholtz Zentrum München - Deutsches Forschungszentrum für Gesundheit und Umwelt
There are currently great hopes for solid-state batteries. They contain no liquid parts that could leak or catch fire. For this reason, they do not require cooling and are considered to be much safer, more reliable, and longer lasting than traditional lithium-ion batteries. Jülich scientists have now introduced a new concept that allows currents up to ten times greater during charging and discharging than previously described in the literature. The improvement was achieved by a “clever” choice of materials with a focus on consistently good compatibility. All components were made from phosphate compounds, which are well matched both chemically and mechanically.
The low current is considered one of the biggest hurdles in the development of solid-state batteries. It is the reason why the batteries take a relatively long...
New design tool automatically creates nanostructure 3D-print templates for user-given colors
Scientists present work at prestigious SIGGRAPH conference
Most of the objects we see are colored by pigments, but using pigments has disadvantages: such colors can fade, industrial pigments are often toxic, and...
Scientists at the University of California, Los Angeles present new research on a curious cosmic phenomenon known as "whistlers" -- very low frequency packets...
Scientists develop first tool to use machine learning methods to compute flow around interactively designable 3D objects. Tool will be presented at this year’s prestigious SIGGRAPH conference.
When engineers or designers want to test the aerodynamic properties of the newly designed shape of a car, airplane, or other object, they would normally model...
Researchers from TU Graz and their industry partners have unveiled a world first: the prototype of a robot-controlled, high-speed combined charging system (CCS) for electric vehicles that enables series charging of cars in various parking positions.
Global demand for electric vehicles is forecast to rise sharply: by 2025, the number of new vehicle registrations is expected to reach 25 million per year....
17.08.2018 | Event News
08.08.2018 | Event News
27.07.2018 | Event News
21.08.2018 | Medical Engineering
21.08.2018 | Physics and Astronomy
21.08.2018 | Power and Electrical Engineering