As NIST researchers reported at a conference on March 20, 2013,* both red and green laser pointers often emitted more visible power than allowed under the Code of Federal Regulations (CFR), and green pointers often emitted unacceptable levels of infrared light as well.
Anecdotal reports of green laser hazards have previously appeared in scientific journals and the media, but the new NIST tests are the first reported precision measurements of a large number of handheld laser devices. The NIST tests point out that many red laser pointers are also—unexpectedly—out of compliance with federal regulations. "Our results raise numerous safety questions regarding laser pointers and their use," the new paper states.
The NIST tests were conducted on randomly selected commercial laser devices labeled as Class IIIa or 3R and sold as suitable for demonstration use in classrooms and other public spaces. Such lasers are limited under the CFR to 5 milliwatts maximum emission in the visible portion of the spectrum and less than 2 milliwatts in the infrared portion of the spectrum. About half the devices tested emitted power levels at least twice the CFR limit at one or more wavelengths. The highest measured power output was 66.5 milliwatts, more than 10 times the legal limit. The power measurements were accurate to within 5 percent.
According to the American National Standards Institute (ANSI), laser devices that exceed 3R limits may be hazardous and should be subject to more rigorous controls such as training, to prevent injury.**
NIST is a non-regulatory agency with decades of experience providing industry, research and military agencies with laser power measurements traceable to international standards. NIST also has a history of innovation in devices for making such measurements. Technical staff from NIST's Laser Radiometry Project built the laser pointer test bed and collaborated with the NIST Office of Safety, Health and Environment on the tests. NIST has provided its data on laser pointer power measurements to the Food and Drug Administration, which regulates laser product safety.
Green lasers generate green light from infrared light. Ideally, the device should be designed and manufactured to confine the infrared light within the laser housing. However, according to the new NIST results, more than 75 percent of the devices tested emitted infrared light in excess of the CFR limit.
NIST Laser Safety Officer Joshua Hadler designed the measurement test bed.*** The system consists of a laser power meter and two optical filters to quantify the emissions of different wavelengths of visible and infrared light. The power meter and filters were calibrated at NIST. Lens holders ensure repeatable laser alignment, and an adjustable aperture contains the laser light around the output end of the laser.
"The measurement system is designed so that anyone can build it using off-the-shelf parts for about $2,000," Hadler says. "By relying on manufacturers' traceability to a national measurement institute such as NIST, someone could use this design to accurately measure power from a laser pointer."* J. Hadler. Random testing reveals excessive power in commercial laser pointers. Presentation at the International Laser Safety Conference, Orlando, Fla., March 20, 2013; J. Hadler, E.L. Tobares and M. Dowell. Random testing reveals excessive power in commercial laser pointers. Journal of Laser Applications. (Forthcoming.)
Laura Ost | EurekAlert!
Physics boosts artificial intelligence methods
19.10.2017 | California Institute of Technology
NASA team finds noxious ice cloud on saturn's moon titan
19.10.2017 | NASA/Goddard Space Flight Center
University of Maryland researchers contribute to historic detection of gravitational waves and light created by event
On August 17, 2017, at 12:41:04 UTC, scientists made the first direct observation of a merger between two neutron stars--the dense, collapsed cores that remain...
Seven new papers describe the first-ever detection of light from a gravitational wave source. The event, caused by two neutron stars colliding and merging together, was dubbed GW170817 because it sent ripples through space-time that reached Earth on 2017 August 17. Around the world, hundreds of excited astronomers mobilized quickly and were able to observe the event using numerous telescopes, providing a wealth of new data.
Previous detections of gravitational waves have all involved the merger of two black holes, a feat that won the 2017 Nobel Prize in Physics earlier this month....
Material defects in end products can quickly result in failures in many areas of industry, and have a massive impact on the safe use of their products. This is why, in the field of quality assurance, intelligent, nondestructive sensor systems play a key role. They allow testing components and parts in a rapid and cost-efficient manner without destroying the actual product or changing its surface. Experts from the Fraunhofer IZFP in Saarbrücken will be presenting two exhibits at the Blechexpo in Stuttgart from 7–10 November 2017 that allow fast, reliable, and automated characterization of materials and detection of defects (Hall 5, Booth 5306).
When quality testing uses time-consuming destructive test methods, it can result in enormous costs due to damaging or destroying the products. And given that...
Using a new cooling technique MPQ scientists succeed at observing collisions in a dense beam of cold and slow dipolar molecules.
How do chemical reactions proceed at extremely low temperatures? The answer requires the investigation of molecular samples that are cold, dense, and slow at...
Scientists from the Max Planck Institute of Quantum Optics, using high precision laser spectroscopy of atomic hydrogen, confirm the surprisingly small value of the proton radius determined from muonic hydrogen.
It was one of the breakthroughs of the year 2010: Laser spectroscopy of muonic hydrogen resulted in a value for the proton charge radius that was significantly...
17.10.2017 | Event News
10.10.2017 | Event News
10.10.2017 | Event News
19.10.2017 | Materials Sciences
19.10.2017 | Materials Sciences
19.10.2017 | Physics and Astronomy