As NIST researchers reported at a conference on March 20, 2013,* both red and green laser pointers often emitted more visible power than allowed under the Code of Federal Regulations (CFR), and green pointers often emitted unacceptable levels of infrared light as well.
Anecdotal reports of green laser hazards have previously appeared in scientific journals and the media, but the new NIST tests are the first reported precision measurements of a large number of handheld laser devices. The NIST tests point out that many red laser pointers are also—unexpectedly—out of compliance with federal regulations. "Our results raise numerous safety questions regarding laser pointers and their use," the new paper states.
The NIST tests were conducted on randomly selected commercial laser devices labeled as Class IIIa or 3R and sold as suitable for demonstration use in classrooms and other public spaces. Such lasers are limited under the CFR to 5 milliwatts maximum emission in the visible portion of the spectrum and less than 2 milliwatts in the infrared portion of the spectrum. About half the devices tested emitted power levels at least twice the CFR limit at one or more wavelengths. The highest measured power output was 66.5 milliwatts, more than 10 times the legal limit. The power measurements were accurate to within 5 percent.
According to the American National Standards Institute (ANSI), laser devices that exceed 3R limits may be hazardous and should be subject to more rigorous controls such as training, to prevent injury.**
NIST is a non-regulatory agency with decades of experience providing industry, research and military agencies with laser power measurements traceable to international standards. NIST also has a history of innovation in devices for making such measurements. Technical staff from NIST's Laser Radiometry Project built the laser pointer test bed and collaborated with the NIST Office of Safety, Health and Environment on the tests. NIST has provided its data on laser pointer power measurements to the Food and Drug Administration, which regulates laser product safety.
Green lasers generate green light from infrared light. Ideally, the device should be designed and manufactured to confine the infrared light within the laser housing. However, according to the new NIST results, more than 75 percent of the devices tested emitted infrared light in excess of the CFR limit.
NIST Laser Safety Officer Joshua Hadler designed the measurement test bed.*** The system consists of a laser power meter and two optical filters to quantify the emissions of different wavelengths of visible and infrared light. The power meter and filters were calibrated at NIST. Lens holders ensure repeatable laser alignment, and an adjustable aperture contains the laser light around the output end of the laser.
"The measurement system is designed so that anyone can build it using off-the-shelf parts for about $2,000," Hadler says. "By relying on manufacturers' traceability to a national measurement institute such as NIST, someone could use this design to accurately measure power from a laser pointer."* J. Hadler. Random testing reveals excessive power in commercial laser pointers. Presentation at the International Laser Safety Conference, Orlando, Fla., March 20, 2013; J. Hadler, E.L. Tobares and M. Dowell. Random testing reveals excessive power in commercial laser pointers. Journal of Laser Applications. (Forthcoming.)
Laura Ost | EurekAlert!
A better way to weigh millions of solitary stars
15.12.2017 | Vanderbilt University
A chip for environmental and health monitoring
15.12.2017 | Friedrich-Alexander-Universität Erlangen-Nürnberg
DNA molecules that follow specific instructions could offer more precise molecular control of synthetic chemical systems, a discovery that opens the door for engineers to create molecular machines with new and complex behaviors.
Researchers have created chemical amplifiers and a chemical oscillator using a systematic method that has the potential to embed sophisticated circuit...
MPQ scientists achieve long storage times for photonic quantum bits which break the lower bound for direct teleportation in a global quantum network.
Concerning the development of quantum memories for the realization of global quantum networks, scientists of the Quantum Dynamics Division led by Professor...
Researchers have developed a water cloaking concept based on electromagnetic forces that could eliminate an object's wake, greatly reducing its drag while...
Tiny pores at a cell's entryway act as miniature bouncers, letting in some electrically charged atoms--ions--but blocking others. Operating as exquisitely sensitive filters, these "ion channels" play a critical role in biological functions such as muscle contraction and the firing of brain cells.
To rapidly transport the right ions through the cell membrane, the tiny channels rely on a complex interplay between the ions and surrounding molecules,...
The miniaturization of the current technology of storage media is hindered by fundamental limits of quantum mechanics. A new approach consists in using so-called spin-crossover molecules as the smallest possible storage unit. Similar to normal hard drives, these special molecules can save information via their magnetic state. A research team from Kiel University has now managed to successfully place a new class of spin-crossover molecules onto a surface and to improve the molecule’s storage capacity. The storage density of conventional hard drives could therefore theoretically be increased by more than one hundred fold. The study has been published in the scientific journal Nano Letters.
Over the past few years, the building blocks of storage media have gotten ever smaller. But further miniaturization of the current technology is hindered by...
11.12.2017 | Event News
08.12.2017 | Event News
07.12.2017 | Event News
15.12.2017 | Trade Fair News
15.12.2017 | Physics and Astronomy
15.12.2017 | Information Technology