As NIST researchers reported at a conference on March 20, 2013,* both red and green laser pointers often emitted more visible power than allowed under the Code of Federal Regulations (CFR), and green pointers often emitted unacceptable levels of infrared light as well.
Anecdotal reports of green laser hazards have previously appeared in scientific journals and the media, but the new NIST tests are the first reported precision measurements of a large number of handheld laser devices. The NIST tests point out that many red laser pointers are also—unexpectedly—out of compliance with federal regulations. "Our results raise numerous safety questions regarding laser pointers and their use," the new paper states.
The NIST tests were conducted on randomly selected commercial laser devices labeled as Class IIIa or 3R and sold as suitable for demonstration use in classrooms and other public spaces. Such lasers are limited under the CFR to 5 milliwatts maximum emission in the visible portion of the spectrum and less than 2 milliwatts in the infrared portion of the spectrum. About half the devices tested emitted power levels at least twice the CFR limit at one or more wavelengths. The highest measured power output was 66.5 milliwatts, more than 10 times the legal limit. The power measurements were accurate to within 5 percent.
According to the American National Standards Institute (ANSI), laser devices that exceed 3R limits may be hazardous and should be subject to more rigorous controls such as training, to prevent injury.**
NIST is a non-regulatory agency with decades of experience providing industry, research and military agencies with laser power measurements traceable to international standards. NIST also has a history of innovation in devices for making such measurements. Technical staff from NIST's Laser Radiometry Project built the laser pointer test bed and collaborated with the NIST Office of Safety, Health and Environment on the tests. NIST has provided its data on laser pointer power measurements to the Food and Drug Administration, which regulates laser product safety.
Green lasers generate green light from infrared light. Ideally, the device should be designed and manufactured to confine the infrared light within the laser housing. However, according to the new NIST results, more than 75 percent of the devices tested emitted infrared light in excess of the CFR limit.
NIST Laser Safety Officer Joshua Hadler designed the measurement test bed.*** The system consists of a laser power meter and two optical filters to quantify the emissions of different wavelengths of visible and infrared light. The power meter and filters were calibrated at NIST. Lens holders ensure repeatable laser alignment, and an adjustable aperture contains the laser light around the output end of the laser.
"The measurement system is designed so that anyone can build it using off-the-shelf parts for about $2,000," Hadler says. "By relying on manufacturers' traceability to a national measurement institute such as NIST, someone could use this design to accurately measure power from a laser pointer."* J. Hadler. Random testing reveals excessive power in commercial laser pointers. Presentation at the International Laser Safety Conference, Orlando, Fla., March 20, 2013; J. Hadler, E.L. Tobares and M. Dowell. Random testing reveals excessive power in commercial laser pointers. Journal of Laser Applications. (Forthcoming.)
Laura Ost | EurekAlert!
'Frequency combs' ID chemicals within the mid-infrared spectral region
16.03.2018 | American Institute of Physics
Fraunhofer HHI have developed a novel single-polarization Kramers-Kronig receiver scheme
16.03.2018 | Fraunhofer-Institut für Nachrichtentechnik, Heinrich-Hertz-Institut, HHI
Animal photoreceptors capture light with photopigments. Researchers from the University of Göttingen have now discovered that these photopigments fulfill an...
On 15 March, the AWI research aeroplane Polar 5 will depart for Greenland. Concentrating on the furthest northeast region of the island, an international team...
The world’s second-largest ice shelf was the destination for a Polarstern expedition that ended in Punta Arenas, Chile on 14th March 2018. Oceanographers from...
At the 2018 ILA Berlin Air Show from April 25–29, the Fraunhofer Institute for Laser Technology ILT is showcasing extreme high-speed Laser Material Deposition (EHLA): A video documents how for metal components that are highly loaded, EHLA has already proved itself as an alternative to hard chrome plating, which is now allowed only under special conditions.
When the EU restricted the use of hexavalent chromium compounds to special applications requiring authorization, the move prompted a rethink in the surface...
At the ILA Berlin, hall 4, booth 202, Fraunhofer FHR will present two radar sensors for navigation support of drones. The sensors are valuable components in the implementation of autonomous flying drones: they function as obstacle detectors to prevent collisions. Radar sensors also operate reliably in restricted visibility, e.g. in foggy or dusty conditions. Due to their ability to measure distances with high precision, the radar sensors can also be used as altimeters when other sources of information such as barometers or GPS are not available or cannot operate optimally.
Drones play an increasingly important role in the area of logistics and services. Well-known logistic companies place great hope in these compact, aerial...
16.03.2018 | Event News
13.03.2018 | Event News
08.03.2018 | Event News
16.03.2018 | Earth Sciences
16.03.2018 | Physics and Astronomy
16.03.2018 | Life Sciences