Nowadays we don't only take measurements with simple measuring devices, but also with whole measuring systems. These are very complex and are completely set up from their component parts at the point of use.
Vehicle scales are an example of such a system. They weigh vehicles like trains, lorries or cars. Unfortunately these complete systems are susceptible to electromagnetic radiation such as that transmitted by mobile phones and radio transceivers – and this is despite the fact that the individual components of the system have passed the standardized test for exactly these types of radiation. This type of test is called an Electromagnetic Compatibility Test and should now also allow the Verification Authorities to test complete systems on site. For this purpose the PTB has developed a facility for use on site, to test electromagnetic compatibility.
The legal requirements for measuring devices subject to legal control also include the EMC, which is tested within the scope of the type examination by PTB. Hereby, the single components of the measuring system are exposed to defined electromagnetic fields in the laboratory. Although the components have passed an EMC test conforming to standards, the measuring systems installed on site are occasionally disturbed by radio receivers or mobile telephones such that false measurement values are shown. For this reason, the responsible authorities have in several cases rejected the verification of vehicle scales.
The awareness that the interference resistance of measuring systems is very decisively dependent on the configuration and the installation on site has not been sufficiently taken into account in the normative requirements. This discrepancy is based on the fact that the European testing requirements worked out several years ago do not sufficiently take into account the actual present-day disturbance source situation due to the spread of radio receivers and mobile telephones.
Due to this technical requirement and also the possible political consequences, a revision of the respective standards was initiated in which the PTB is participating. For the determination of new normative limiting values and for the assessment of the interference resistance of measuring devices on site by the verification authorities, metrologically traceable EMC tests on site are necessary, for which there has not been a measuring device available up to now.
For this reason, a transportable testing device was developed at the PTB, which enables testing at discrete frequencies between 27 MHz and 5.8 GHz. The frequencies were selected such that, on the one hand, the real disturbance sources are displayed and, on the other hand, the disturbance of radio services is avoided, so that as a result, the Federal Network Agency could grant a special permit for the restricted test operation. The frequencies of the testing device lie in the vicinity of the frequency bands of classical means of communication such as company radio and amateur radio, but also in the vicinity of the frequency ranges of modern communications systems such as GSM, DECT, UMTS, Bluetooth and WLAN.
In the test operation on site, the critical parts of the measuring system can at a distance of 1 m from the transmitting antenna be charged with the electromagnetic field successively for 1 min in each case for each single frequency and thereby the correct function of the measuring system be assessed. By means of a suitable design and a software user guidance, the presence of an expert in high frequency engineering is not necessary for the operation of the testing device on site.
Imke Frischmuth | alfa
SF State astronomer searches for signs of life on Wolf 1061 exoplanet
20.01.2017 | San Francisco State University
Molecule flash mob
19.01.2017 | Technische Universität Wien
An important step towards a completely new experimental access to quantum physics has been made at University of Konstanz. The team of scientists headed by...
Yersiniae cause severe intestinal infections. Studies using Yersinia pseudotuberculosis as a model organism aim to elucidate the infection mechanisms of these...
Researchers from the University of Hamburg in Germany, in collaboration with colleagues from the University of Aarhus in Denmark, have synthesized a new superconducting material by growing a few layers of an antiferromagnetic transition-metal chalcogenide on a bismuth-based topological insulator, both being non-superconducting materials.
While superconductivity and magnetism are generally believed to be mutually exclusive, surprisingly, in this new material, superconducting correlations...
Laser-driving of semimetals allows creating novel quasiparticle states within condensed matter systems and switching between different states on ultrafast time scales
Studying properties of fundamental particles in condensed matter systems is a promising approach to quantum field theory. Quasiparticles offer the opportunity...
Among the general public, solar thermal energy is currently associated with dark blue, rectangular collectors on building roofs. Technologies are needed for aesthetically high quality architecture which offer the architect more room for manoeuvre when it comes to low- and plus-energy buildings. With the “ArKol” project, researchers at Fraunhofer ISE together with partners are currently developing two façade collectors for solar thermal energy generation, which permit a high degree of design flexibility: a strip collector for opaque façade sections and a solar thermal blind for transparent sections. The current state of the two developments will be presented at the BAU 2017 trade fair.
As part of the “ArKol – development of architecturally highly integrated façade collectors with heat pipes” project, Fraunhofer ISE together with its partners...
19.01.2017 | Event News
10.01.2017 | Event News
09.01.2017 | Event News
20.01.2017 | Awards Funding
20.01.2017 | Materials Sciences
20.01.2017 | Life Sciences