Nobel Direct was launched in 2004 by Nobel Biocare AB. The implants were considered a great innovation, as they could be screwed directly into the jawbone without having to first lift up the mucous membrane.
“We have followed up 48 patients who were among the first to get the implant. For each passing year, we have been able to see how the problems related to these implants have grown more and more,” observes Pär-Olov Östman, a dentist who will be presented the study tomorrow at the defence of his dissertation.
After one year, about five per cent of the implants had been lost, and 20 per cent of the remaining implants showed bone loss of more than three millimetres. The new report indicates that after three years, eight per cent of the implants had been lost, and 25 per cent of the remaining implants showed bone loss of more than three millimetres.
“We believe that the problems related to Nobel Direct result both from the design of the implant and an uneven surface against the soft tissue in combination with the method of treatment recommended by the company,” according to Professor Lars Sennerby.
For some time, Nobel Direct was marketed as an implant that was easy to use, and therefore suitable for less experienced dentists. According to the company’s marketing, the implant would also counteract marginal bone loss.
”If the implant is inserted in a more conservative manner, avoiding direct load, the results appear to be better. We believe that there are additional implants with similar design that can also cause problems if they are inserted in the same way as Nobel Direct. However, we are unable to draw any certain conclusions regarding these,” says Professor Tomas Albrektsson, the head of the Department for Biomaterials Science at the Sahlgrenska Academy.
At the request of the Medical Products Agency, Nobel Biocare AB is now working to clarify certain information in the product information material. The company also markets several other titanium implants that have been scientifically proved to be very safe, including a Brånemark implant with the same patented surface as the Nobel Direct implant in dispute.FACTS ON DENTAL IMPLANTAT
Ulrika Lundin | alfa
New malaria analysis method reveals disease severity in minutes
14.08.2017 | University of British Columbia
New type of blood cells work as indicators of autoimmunity
14.08.2017 | Instituto de Medicina Molecular
16.08.2017 | Event News
04.08.2017 | Event News
26.07.2017 | Event News
21.08.2017 | Medical Engineering
21.08.2017 | Materials Sciences
21.08.2017 | Life Sciences