However, it isn’t that simple for researchers who need to measure temperatures in microfluidic systems—tiny, channel-lined devices used in medical diagnostics, DNA forensics and “lab-on-a-chip” chemical analyzers—as their current “thermometer” can only be precisely calibrated for one reference temperature.
Now, researchers at the National Institute of Standards and Technology (NIST) have proposed a mathematical solution that enables researchers to calibrate the “thermometer” for microfluidic systems so that all temperatures are covered.
Reactions taking place in microfluidic systems often require heating, meaning that users must accurately monitor temperature changes in fluid volumes ranging from a few microliters (a droplet approximately 1 millimeter in diameter) to sub-nanoliters (a droplet approximately 1/10 of millimeter in diameter). A common DNA analysis technique, for example, depends heavily on precise temperature cycling.
Ordinary thermometers or other temperature probes are useless at such tiny dimensions, so some groups have turned to temperature-sensitive fluorescent dyes, particularly rhodamine B. The intensity of the dye’s fluorescence decreases with increasing temperature. The idea is that the dye can be used as a noninvasive way to map the range of temperatures occurring within a microfluidic system during heating and, in turn, provide a means of calibrating that system for experiments.
However, the technique currently requires the user to base all readings on the fluorescence at a single reference temperature. Previous groups have developed “calibration curves” that relate temperature to rhodmaine B fluorescent intensity based on a reference temperature of about 23 degrees Celsius (a technique first proposed by NIST researchers David Ross, Michael Gaitan and Laurie Locascio in 2001*). But it turns out that the curves are only good for that one temperature. In an upcoming paper in Analytical Chemistry**, the NIST team—Jayna J. Shah, Michael Gaitan and Jon Geist—reports that changing the reference point, such as the higher temperature when a microfluidic system is first heated, introduces errors when a dye intensity-to-temperature calculation is done using current methods.
“Our analysis shows that a simple linear correction for a 40 degrees Celsius reference temperature identified errors between minus 3 to 8 degrees Celsius for three previously published sets of calibration equations derived at approximately 23 degrees Celsius,” says lead researcher Shah.
To address the problem, the NIST team developed mathematical methods to correct for the shift experienced when the reference temperature changes. This allowed the researchers to create generalized calibration equations that can be applied to any reference temperature.
Microfluidic DNA amplification (production of numerous copies of DNA from a tiny sample) by the polymerase chain reaction (PCR) is one procedure that could benefit from the new NIST calculations, Shah says. “PCR requires a microfluidic device to be cycled through temperatures at three different zones starting around 65 degrees Celsius, so a useful dye intensity-to-temperature ratio would have to be based on that temperature and not a reference point of 23 degrees Celsius,” she explains.
* D. Ross, M. Gaitan and L.E. Locascio. Temperature measurement in microfluidic systems using a temperature-dependent fluorescent dye. Analytical Chemistry, Vol. 73, No. 17, pages 4117-4123, Sept. 1, 2001.
** J.J. Shah, M. Gaitan and J. Geist. Generalized temperature measurement equations for rhodamine B dye solution and its application to microfluidics. Analytical Chemistry, Vol. 81, No. 19, Oct. 1, 2009 (published online Sept. 1, 2009).
Michael E. Newman | Newswise Science News
New risk factors for anxiety disorders
24.02.2017 | Julius-Maximilians-Universität Würzburg
Stingless bees have their nests protected by soldiers
24.02.2017 | Johannes Gutenberg-Universität Mainz
In the field of nanoscience, an international team of physicists with participants from Konstanz has achieved a breakthrough in understanding heat transport
Cells need to repair damaged DNA in our genes to prevent the development of cancer and other diseases. Our cells therefore activate and send “repair-proteins”...
The Fraunhofer IWS Dresden and Technische Universität Dresden inaugurated their jointly operated Center for Additive Manufacturing Dresden (AMCD) with a festive ceremony on February 7, 2017. Scientists from various disciplines perform research on materials, additive manufacturing processes and innovative technologies, which build up components in a layer by layer process. This technology opens up new horizons for component design and combinations of functions. For example during fabrication, electrical conductors and sensors are already able to be additively manufactured into components. They provide information about stress conditions of a product during operation.
The 3D-printing technology, or additive manufacturing as it is often called, has long made the step out of scientific research laboratories into industrial...
Nature does amazing things with limited design materials. Grass, for example, can support its own weight, resist strong wind loads, and recover after being...
Nanometer-scale magnetic perforated grids could create new possibilities for computing. Together with international colleagues, scientists from the Helmholtz Zentrum Dresden-Rossendorf (HZDR) have shown how a cobalt grid can be reliably programmed at room temperature. In addition they discovered that for every hole ("antidot") three magnetic states can be configured. The results have been published in the journal "Scientific Reports".
Physicist Dr. Rantej Bali from the HZDR, together with scientists from Singapore and Australia, designed a special grid structure in a thin layer of cobalt in...
13.02.2017 | Event News
10.02.2017 | Event News
09.02.2017 | Event News
24.02.2017 | Life Sciences
24.02.2017 | Life Sciences
24.02.2017 | Trade Fair News