"This clinical trial has validated that the Tongue Drive system is intuitive and quite simple for individuals with high-level spinal cord injuries to use," said Maysam Ghovanloo, an assistant professor in the School of Electrical and Computer Engineering at the Georgia Institute of Technology.
"Trial participants were able to easily remember and correctly issue tongue commands to play computer games and drive a powered wheelchair around an obstacle course with very little prior training."At the annual conference of the Rehabilitation Engineering and Assistive Technology Society of North America (RESNA) on June 26, the researchers reported the results of the first five clinical trial subjects to use the Tongue Drive system. The trial was conducted at the Shepherd Center, an Atlanta-based catastrophic care hospital, and funded by the National Science Foundation and the Christopher and Dana Reeve Foundation.
At the beginning of each trial, Ghovanloo and graduate students Xueliang Huo and Chih-wen Cheng attached a small magnet -- the size of a grain of rice -- to the participant's tongue with tissue adhesive. Movement of this magnetic tracer was detected by an array of magnetic field sensors mounted on wireless headphones worn by the subject. The sensor output signals were wirelessly transmitted to a portable computer, which was carried on the wheelchair.The signals were processed to determine the relative motion of the magnet with respect to the array of sensors in real-time. This information was then used to control the movements of the cursor on a computer screen or to substitute for the joystick function in a powered wheelchair. Details on use of the Tongue Drive for wheeled mobility were published in the June 2009 issue of the journal IEEE Transactions on Biomedical Engineering.
Before using the Tongue Drive system, the subjects trained the computer to understand how they would like to move their tongues to indicate different commands. A unique set of specific tongue movements was tailored for each individual based on the user's abilities, oral anatomy and personal preferences. For the first computer test, the user issued commands to move the computer mouse left and right. Using these commands, each subject played a computer game that required moving a paddle horizontally to prevent a ball from hitting the bottom of the screen.
After adding two more commands to their repertoire -- up and down -- the subjects were asked to move the mouse cursor through an on-screen maze as quickly and accurately as possible.
Then the researchers added two more commands -- single and double mouse clicks -- to provide the subject with complete mouse functionality. When a randomly selected symbol representing one of the six commands appeared on the computer screen, the subject was instructed to issue that command within a specified time period. Each subject completed 40 trials for each time period.
After the computer sessions, the subjects were ready for the wheelchair driving exercise. Using forward, backward, right, left and stop/neutral tongue commands, the subjects maneuvered a powered wheelchair through an obstacle course.
The obstacle course contained 10 turns and was longer than a professional basketball court. Throughout the course, the users had to perform navigation tasks such as making a U-turn, backing up and fine-tuning the direction of the wheelchair in a limited space. Subjects were asked to navigate through the course as fast as they could, while avoiding collisions.
Each subject operated the powered wheelchair using two different control strategies: discrete mode, which was designed for novice users, and continuous mode for more experienced users. In discrete mode, if the user issued the command to move forward and then wanted to turn right, the user would have to stop the wheelchair before issuing the command to turn right. The stop command was selected automatically when the tongue returned to its resting position, bringing the wheelchair to a standstill.
"Discrete mode is a safety feature particularly for novice users, but it reduces the agility of the wheelchair movement," explained Ghovanloo. "In continuous mode, however, the user is allowed to steer the powered wheelchair to the left or right as it is moving forward and backward, thus making it possible to follow a curve."
Each subject completed the course at least twice using each strategy while the researchers recorded the navigation time and number of collisions. Using discrete control, the average speed for the five subjects was 5.2 meters per minute and the average number of collisions was 1.8. Using continuous control, the average speed was 7.7 meters per minute and the average number of collisions was 2.5.
While this initial performance trial only required six tongue commands, the Tongue Drive system can potentially capture a large number of tongue movements, each of which can represent a different user command. The ability to train the system with as many commands as an individual can comfortably remember and having all of the commands available to the user at the same time are significant advantages over the common sip-n-puff device that acts as a simple switch controlled by sucking or blowing through a straw.
Some sip-n-puff users also consider the straw to be a symbol of their disability. Since Tongue Drive users simply wear headphones that are commonly worn to listen to music, the system is more acceptable to potential users.
John Anschutz, manager of the assistive technology program at the Shepherd Center, identified advantages the Tongue Drive system has over the tongue-touch keypad.
"The Tongue Drive system seems to be much more supportable if there were a failure of some component within the system. With the old tongue-touch keypad, if the system went down then the user lost all of the functions of the wheelchair, phone, computer and environmental control," explained Anschutz. "Ghovanloo's approach should be much more repairable should a fault arise, which is critical for systems for which so much function is depended upon."
A future system upgrade will be to move the sensors inside the user's mouth, according to Ghovanloo. This will be an important step for users who are very impaired and cannot reposition the system for best results, according to Anschutz.
"All of the subjects successfully completed the computer and powered wheelchair navigation tasks with their tongues without difficulty, which demonstrates that the Tongue Drive system can potentially provide individuals unable to move their arms and hands with effective control over a wide variety of devices they use in their daily lives," said Ghovanloo.
Abby Vogel | EurekAlert!
'Super yeast' has the power to improve economics of biofuels
18.10.2016 | University of Wisconsin-Madison
Engineers reveal fabrication process for revolutionary transparent sensors
14.10.2016 | University of Wisconsin-Madison
Researchers from the Institute for Quantum Computing (IQC) at the University of Waterloo led the development of a new extensible wiring technique capable of controlling superconducting quantum bits, representing a significant step towards to the realization of a scalable quantum computer.
"The quantum socket is a wiring method that uses three-dimensional wires based on spring-loaded pins to address individual qubits," said Jeremy Béjanin, a PhD...
In a paper in Scientific Reports, a research team at Worcester Polytechnic Institute describes a novel light-activated phenomenon that could become the basis for applications as diverse as microscopic robotic grippers and more efficient solar cells.
A research team at Worcester Polytechnic Institute (WPI) has developed a revolutionary, light-activated semiconductor nanocomposite material that can be used...
By forcefully embedding two silicon atoms in a diamond matrix, Sandia researchers have demonstrated for the first time on a single chip all the components needed to create a quantum bridge to link quantum computers together.
"People have already built small quantum computers," says Sandia researcher Ryan Camacho. "Maybe the first useful one won't be a single giant quantum computer...
COMPAMED has become the leading international marketplace for suppliers of medical manufacturing. The trade fair, which takes place every November and is co-located to MEDICA in Dusseldorf, has been steadily growing over the past years and shows that medical technology remains a rapidly growing market.
In 2016, the joint pavilion by the IVAM Microtechnology Network, the Product Market “High-tech for Medical Devices”, will be located in Hall 8a again and will...
'Ferroelectric' materials can switch between different states of electrical polarization in response to an external electric field. This flexibility means they show promise for many applications, for example in electronic devices and computer memory. Current ferroelectric materials are highly valued for their thermal and chemical stability and rapid electro-mechanical responses, but creating a material that is scalable down to the tiny sizes needed for technologies like silicon-based semiconductors (Si-based CMOS) has proven challenging.
Now, Hiroshi Funakubo and co-workers at the Tokyo Institute of Technology, in collaboration with researchers across Japan, have conducted experiments to...
14.10.2016 | Event News
14.10.2016 | Event News
12.10.2016 | Event News
21.10.2016 | Health and Medicine
21.10.2016 | Information Technology
21.10.2016 | Materials Sciences