"This clinical trial has validated that the Tongue Drive system is intuitive and quite simple for individuals with high-level spinal cord injuries to use," said Maysam Ghovanloo, an assistant professor in the School of Electrical and Computer Engineering at the Georgia Institute of Technology.
"Trial participants were able to easily remember and correctly issue tongue commands to play computer games and drive a powered wheelchair around an obstacle course with very little prior training."At the annual conference of the Rehabilitation Engineering and Assistive Technology Society of North America (RESNA) on June 26, the researchers reported the results of the first five clinical trial subjects to use the Tongue Drive system. The trial was conducted at the Shepherd Center, an Atlanta-based catastrophic care hospital, and funded by the National Science Foundation and the Christopher and Dana Reeve Foundation.
At the beginning of each trial, Ghovanloo and graduate students Xueliang Huo and Chih-wen Cheng attached a small magnet -- the size of a grain of rice -- to the participant's tongue with tissue adhesive. Movement of this magnetic tracer was detected by an array of magnetic field sensors mounted on wireless headphones worn by the subject. The sensor output signals were wirelessly transmitted to a portable computer, which was carried on the wheelchair.The signals were processed to determine the relative motion of the magnet with respect to the array of sensors in real-time. This information was then used to control the movements of the cursor on a computer screen or to substitute for the joystick function in a powered wheelchair. Details on use of the Tongue Drive for wheeled mobility were published in the June 2009 issue of the journal IEEE Transactions on Biomedical Engineering.
Before using the Tongue Drive system, the subjects trained the computer to understand how they would like to move their tongues to indicate different commands. A unique set of specific tongue movements was tailored for each individual based on the user's abilities, oral anatomy and personal preferences. For the first computer test, the user issued commands to move the computer mouse left and right. Using these commands, each subject played a computer game that required moving a paddle horizontally to prevent a ball from hitting the bottom of the screen.
After adding two more commands to their repertoire -- up and down -- the subjects were asked to move the mouse cursor through an on-screen maze as quickly and accurately as possible.
Then the researchers added two more commands -- single and double mouse clicks -- to provide the subject with complete mouse functionality. When a randomly selected symbol representing one of the six commands appeared on the computer screen, the subject was instructed to issue that command within a specified time period. Each subject completed 40 trials for each time period.
After the computer sessions, the subjects were ready for the wheelchair driving exercise. Using forward, backward, right, left and stop/neutral tongue commands, the subjects maneuvered a powered wheelchair through an obstacle course.
The obstacle course contained 10 turns and was longer than a professional basketball court. Throughout the course, the users had to perform navigation tasks such as making a U-turn, backing up and fine-tuning the direction of the wheelchair in a limited space. Subjects were asked to navigate through the course as fast as they could, while avoiding collisions.
Each subject operated the powered wheelchair using two different control strategies: discrete mode, which was designed for novice users, and continuous mode for more experienced users. In discrete mode, if the user issued the command to move forward and then wanted to turn right, the user would have to stop the wheelchair before issuing the command to turn right. The stop command was selected automatically when the tongue returned to its resting position, bringing the wheelchair to a standstill.
"Discrete mode is a safety feature particularly for novice users, but it reduces the agility of the wheelchair movement," explained Ghovanloo. "In continuous mode, however, the user is allowed to steer the powered wheelchair to the left or right as it is moving forward and backward, thus making it possible to follow a curve."
Each subject completed the course at least twice using each strategy while the researchers recorded the navigation time and number of collisions. Using discrete control, the average speed for the five subjects was 5.2 meters per minute and the average number of collisions was 1.8. Using continuous control, the average speed was 7.7 meters per minute and the average number of collisions was 2.5.
While this initial performance trial only required six tongue commands, the Tongue Drive system can potentially capture a large number of tongue movements, each of which can represent a different user command. The ability to train the system with as many commands as an individual can comfortably remember and having all of the commands available to the user at the same time are significant advantages over the common sip-n-puff device that acts as a simple switch controlled by sucking or blowing through a straw.
Some sip-n-puff users also consider the straw to be a symbol of their disability. Since Tongue Drive users simply wear headphones that are commonly worn to listen to music, the system is more acceptable to potential users.
John Anschutz, manager of the assistive technology program at the Shepherd Center, identified advantages the Tongue Drive system has over the tongue-touch keypad.
"The Tongue Drive system seems to be much more supportable if there were a failure of some component within the system. With the old tongue-touch keypad, if the system went down then the user lost all of the functions of the wheelchair, phone, computer and environmental control," explained Anschutz. "Ghovanloo's approach should be much more repairable should a fault arise, which is critical for systems for which so much function is depended upon."
A future system upgrade will be to move the sensors inside the user's mouth, according to Ghovanloo. This will be an important step for users who are very impaired and cannot reposition the system for best results, according to Anschutz.
"All of the subjects successfully completed the computer and powered wheelchair navigation tasks with their tongues without difficulty, which demonstrates that the Tongue Drive system can potentially provide individuals unable to move their arms and hands with effective control over a wide variety of devices they use in their daily lives," said Ghovanloo.
Abby Vogel | EurekAlert!
Did you know that the wrapping of Easter eggs benefits from specialty light sources?
13.04.2017 | Heraeus Noblelight GmbH
To e-, or not to e-, the question for the exotic 'Si-III' phase of silicon
05.04.2017 | Carnegie Institution for Science
More and more automobile companies are focusing on body parts made of carbon fiber reinforced plastics (CFRP). However, manufacturing and repair costs must be further reduced in order to make CFRP more economical in use. Together with the Volkswagen AG and five other partners in the project HolQueSt 3D, the Laser Zentrum Hannover e.V. (LZH) has developed laser processes for the automatic trimming, drilling and repair of three-dimensional components.
Automated manufacturing processes are the basis for ultimately establishing the series production of CFRP components. In the project HolQueSt 3D, the LZH has...
Reflecting the structure of composites found in nature and the ancient world, researchers at the University of Illinois at Urbana-Champaign have synthesized thin carbon nanotube (CNT) textiles that exhibit both high electrical conductivity and a level of toughness that is about fifty times higher than copper films, currently used in electronics.
"The structural robustness of thin metal films has significant importance for the reliable operation of smart skin and flexible electronics including...
The nearby, giant radio galaxy M87 hosts a supermassive black hole (BH) and is well-known for its bright jet dominating the spectrum over ten orders of magnitude in frequency. Due to its proximity, jet prominence, and the large black hole mass, M87 is the best laboratory for investigating the formation, acceleration, and collimation of relativistic jets. A research team led by Silke Britzen from the Max Planck Institute for Radio Astronomy in Bonn, Germany, has found strong indication for turbulent processes connecting the accretion disk and the jet of that galaxy providing insights into the longstanding problem of the origin of astrophysical jets.
Supermassive black holes form some of the most enigmatic phenomena in astrophysics. Their enormous energy output is supposed to be generated by the...
The probability to find a certain number of photons inside a laser pulse usually corresponds to a classical distribution of independent events, the so-called...
Microprocessors based on atomically thin materials hold the promise of the evolution of traditional processors as well as new applications in the field of flexible electronics. Now, a TU Wien research team led by Thomas Müller has made a breakthrough in this field as part of an ongoing research project.
Two-dimensional materials, or 2D materials for short, are extremely versatile, although – or often more precisely because – they are made up of just one or a...
28.04.2017 | Event News
20.04.2017 | Event News
18.04.2017 | Event News
28.04.2017 | Medical Engineering
28.04.2017 | Earth Sciences
28.04.2017 | Life Sciences